Wall mounting dental X-ray machine WWG-WX88
Power requirement: 50Hz, 220V±10%
Peak point voltage: 65 kvp
Peak point current: 1.5mA/3mA/7mA/10/mA
Maximum power: ≤ 300w
Exposure time: 0.1-0.2s
Radiation diameter: 6cm
Focus spot: 0.6 mm×0.8 mm
X-ray tube longitudinal displacement:≥ 1600mm
Vertical travel of rocker: ≥ 600mm
1. Filming in coordination with dental film with intensifying screen by making use of 1.5mA X-ray
output capacity. Compared with 7mA-10mA dental unit in the world, X-ray radiation dosage has
reduced by 92%-98% to protect health of patients.
2. The scattering or leaking x-ray dosage in the air where one meter away from radius of
shining center is lower than 4% of defined limited value to protect health of doctors.
3. The clearest image can be acquired by making use of microfocus x-ray tube.
4. Using microcomputer controller, which is very convenient and accurate. Only according to
relative standard picture, the best exposure control can be reached.
5. It has infrared remote control and linear control and the direction and anti-interference of
infrared remote control are strong.
6. It has flexible and firm suspending arm to adjust ball tube freely to any angle or direction and
balance in any position.
There are no reviews yet.